1. How to use the air sterilizer
1. The department keeps the instructions for the disinfection machine and operates as required. 2. Pay attention to the airtightness of the room. Doors and windows should be closed during disinfection to maintain good airtightness in the room. Irrelevant personnel are strictly prohibited from entering, and the number of indoor personnel should be minimized to ensure the disinfection effect. 3. Pay attention to the hygiene of indoor surfaces. This machine is only effective for air and has no disinfection effect on the surface of objects. If there is a lot of dust on the surface of indoor objects, secondary dust will be generated when the disinfection machine is working, causing continuous air pollution by microorganisms, and ultimately leading to disinfection failure within the specified time. 4. Selection of disinfection boot time. 1) Preventive disinfection: regular disinfection 2-3 times a day, 60~120min each time. It is generally arranged before work in the morning and after get off work in the afternoon or at night. The general disinfection time is 5 hours. 2) Dynamic disinfection: the purpose is to control and reduce the secondary pollution of ambient air during personnel activities. It is usually carried out during the peak period of personnel activities. 3) Static disinfection can be carried out once a week for 2 hours. 5. Corresponding records should be made at the end of each disinfection, and the cumulative time should not exceed 4000 hours. 6. The disinfection machine adopts the principle of physical filtration and disinfection of ultraviolet circulating wind, so the space around the disinfection machine should be avoided as far as possible to keep the air flowing smoothly, and the good circulation of the air should be protected as much as possible. 2. Cleaning and maintenance of the disinfection machine 1. Keep the disinfection machine clean and dry. Wipe the surface with a damp cloth after disinfection every day. When cleaning, cut off the power and pull out the power plug to avoid direct contact with water or flushing. 2. When the disinfection machine is working, it is strictly forbidden to let objects or hands approach the ventilation inlet and outlet of the disinfection machine; when handling and loading and unloading, prevent the product from being hit by hard objects or falling to the ground. 3. When the disinfection machine is found to work abnormally (fault display or alarm), turn off the power immediately. If you want to know more information, please call the 24-hour hotline: 18005333355 Manager Zhong Address: Yumin Industrial Park, Zibo High-tech Zone, Shandong
To that end, Qingdao Funglan Environmental Protection & Technology Co., Ltd. has successfully built a solid foundation and infrastructure for air sterilizer manufacturing.
You can always ask the experts for the most appropriate custom air cleaners air sterilizer for your specific needs. Find the best support, prices and other to air sterilizer solutions at Funglan Air Purifier.
Many homeowners find that they can cut costs while keeping home cool efficiently with .