Monitoring of medical air sterilizer
in Xiamen, Fujian-Comparison before and after use of air disinfection machine in Xiamen, Fujian. First of all, thanks to the majority of Fujian Xiamen hospital consumers for purchasing and using manufacturer disinfection products. To operate the sterilizer, Fujian Xiamen Hospital should also maintain and maintain the sterilizer in accordance with the 'Instruction Manual'. After all these tasks were carried out smoothly, many users of Xiamen Hospital in Fujian began to consider how to do a good job in monitoring the medical air sterilizer. The following introduces the monitoring method of medical air disinfection machine◆ , the manufacturer provides better service to many users. 1. Medical air disinfection machine ◆ sterilization factor life monitoring: first calibrate the medical air disinfection machine ◆ Sterilization factor working timer, the average effective life of the lamp is 8000h, according to the requirements of the air disinfection machine product 'Instruction Manual' The lamp accumulates to 7000-8000h, and the irradiance value of the lamp is measured. The users of Xiamen Hospital in Fujian should replace the unqualified lamp in time. 2. Medical air disinfection machine ◆ Measurement of the irradiance value of the lamp tube: Use a special lamp irradiometer to monitor the lamp tube of the air disinfection machine. 3. Monitoring of the air disinfection effect of the air disinfection machine@ medical air disinfection machine: ①. Sampling time * Close the doors and windows of the disinfection room, and take samples after the disinfection of the air disinfection machine. ② Settlement method* Sampling height: 0.8 meters above the ground. When sampling at Xiamen Hospital, Fujian, open the lid of the plate and buckle it upside down next to the plate. Exposure time of the plate: 15 minutes, and finally cover the plate and send it in time Check. ③. Point layout method * If the disinfection room area of u200bu200bFujian Xiamen Hospital is less than or equal to 30 square meters, set three points (inside, middle and outside) on the diagonal, and the (inside, outside) point should be 1 meter away from the wall. The area of u200bu200bdisinfection room in Xiamen Hospital in Fujian is more than 30 square meters. There are five sampling points at the four corners and the center. The sampling points arranged at the four corners should be 1 meter away from the wall. ④. Judgment of results * Medical air disinfection machine◆ After the air sterilizer is disinfected, the total number of bacterial colonies in the air ≤ 4CFU/(15 minutes, 9cm diameter plate) is qualified for disinfection. (Evaluate the disinfection effect of the medical air disinfection machine◆ The air disinfection machine can also take samples before and after air disinfection, and compare the extinction rate of natural bacteria in the air before and after disinfection%) ⑤. Medical air disinfection machine◆ Air Precautions for monitoring the air disinfection effect of the disinfection machine * Close the doors and windows of the disinfection room before sampling in Xiamen Hospital, Fujian Province, and perform sampling at a standstill for 10 minutes when no one is moving. The sampling height and arrangement method before and after air disinfection It is exactly the same, the disinfection time is 2-3h, and the applicable area meets the requirements of the 'Instruction Manual' of the medical air disinfection machine.
air sterilizer are among the best and the long known , which plays an essential part in automatic manufacturing.
Qingdao Funglan Environmental Protection & Technology Co., Ltd. will deliver superior returns to our shareholders by tirelessly pursuing new growth opportunities while continually improving our profitability, a socially responsible, ethical company that is watched and emulated as a model of success.
Qingdao Funglan Environmental Protection & Technology Co., Ltd. can reassign work or shuffle around assigned tasks if one team member is overwhelmed while others are not, more effectively managing resources on the fly. With detailed overviews and reports, manufacturers also can more easily stay abreast of new developments.
The major classifications of are custom air cleaners, custom air cleaners, custom air cleaners and custom air cleaners machines.