How to do the monitoring of medical air sterilizer
? The editor of medical air disinfection machine will introduce the monitoring method of medical air disinfection machine. 1. Lifetime monitoring of sterilization factor: firstly calibrate the sterilization factor working timer of the sterilizer. The average effective life of the ultraviolet lamp is 8000h. According to the product manual of the sterilizer, when the timer accumulates to 7000-8000h, measure the radiation of the ultraviolet lamp. The illuminance value should be replaced in time for the unqualified UV lamps. 2. Monitoring the irradiance value of the ultraviolet lamp: use a special ultraviolet lamp irradiometer to test the ultraviolet lamp. Monitoring method of medical air sterilizer 3. Monitoring the air disinfection effect of medical air disinfection machine: ①, close the doors and windows of the disinfection room of the medical air disinfection machine manufacturer, and take samples after disinfection by the air disinfection machine. ②. At 0.8 meters from the ground, open the lid of the plate and buckle it upside down next to the plate when sampling. The exposure time of the plate: 15 minutes. Finally, cover the plate and send it for inspection in time. ③. If the area of u200bu200bthe disinfection room is less than or equal to 30 square meters, set three points on the diagonal (inside, middle and outside), and the (inside, outside) point should be 1 meter away from the wall. The area of u200bu200bthe disinfection room is more than 30 square meters, and there are five sampling points in the four corners and the center. The sampling points arranged in the four corners should be 1 meter away from the wall. ④. After disinfection, the total number of bacterial colonies in the air ≤4 CFU/(15 minutes, 9cm diameter plate) is qualified for disinfection. (To evaluate the disinfection effect, you can also take samples before and after air disinfection, and compare the extinction rate of natural bacteria in the air before and after disinfection in %). ⑤ Precautions for monitoring the air disinfection effect of medical air disinfection machines: close the doors and windows of the disinfection room before sampling, and take samples at a standstill for 10 minutes when no one is moving. The sampling height and placement before and after air disinfection The method is exactly the same, the disinfection time is 2-3h, and the applicable area meets the requirements of the 'Instruction for Use' of the air disinfector. Recommended reading: 1. brand medical manufacturer 2. How to use medical air disinfection machine 3. Medical air disinfection machine price 4. Plasma medical air disinfection machine which is good 5. What is the medical mobile air disinfection machine
Qingdao Funglan Environmental Protection & Technology Co., Ltd. has various branches in different countries worldwide.
Qingdao Funglan Environmental Protection & Technology Co., Ltd. attaches great importance to customers and assists them in achieving their demands.
The group's Quality Systems Manager (QSM) is responsible for ensuring that Qingdao Funglan Environmental Protection & Technology Co., Ltd. has in place systems that guarantee quality throughout the Group.
There are multiple advantages of having a custom air cleaners air sterilizer from responsible drilling machine exporters such as Qingdao Funglan Environmental Protection & Technology Co., Ltd., as they adhere to all the quality standards as you can list and supply all custom air cleaners essential for the operation of the device without any difficulty.
Diversifying is an excellent growth strategy, as it allows Funglan to have multiple streams of income that can often fill seasonal voids and, of course, increase sales and profit margins.