Urumqi, Xinjiang Medical Air Sterilizer│Medical Air Sterilizer-Troubleshooting Method Air sterilizer is a necessary air purification and disinfection equipment in hospitals, and it is essential to improve the quality of the hospital’s air environment. Hospital operating rooms, ICUs, neonatal rooms, delivery rooms, supply rooms, treatment centers, hemodialysis rooms, infusion rooms, sterile rooms and other departments generally purchase and use medical air sterilizer
s. Few small faults and faults of one kind or another will appear. The average user also feels that these small faults and small faults will be repaired by the manufacturer as if it is unnecessary. How can users quickly deal with these small faults? It is a medical air sterilizer. Basic skills necessary for users. Today’s engineer also gave a video guide to a hospital user in Urumqi, Xinjiang, and now introduces some common troubleshooting methods for air sterilizers. More air sterilizers and medical air sterilizers can be viewed: Fault 1: The air output of the air sterilizer is small or the noise of the sterilizer is too loud. Troubleshooting: the filter of the sterilizer is dusty, remove the filter of the sterilizer to clean or replace the filter. Fault 2: The LCD screen of the medical air disinfection machine warns to clean or replace the filter. Troubleshooting: The cumulative filter cleaning or filter replacement time has reached the alarm time, clean or replace the filter. Fault 3: The LCD screen of the medical air sterilizer warns to replace the high-pressure plasma. Troubleshooting: The accumulative replacement time of the high-pressure plasma reaches the alarm time, and the high-pressure plasma is replaced. Fault 4: The LCD screen of the air disinfector warns to replace the ultraviolet lamp. Troubleshooting: The cumulative replacement time of the ultraviolet lamp has reached the alarm time, and the ultraviolet lamp should be replaced. Fault 5: The air sterilizer does not work, and the power indicator does not light up. Troubleshooting: The power plug of the disinfection machine is in poor contact, replace the power socket. The fuse of the disinfection machine is blown, replace the spare fuse of the disinfection machine. Fault 6: The remote control of the medical air disinfection machine is not sensitive. Troubleshooting: Replace the battery of the remote control of the sterilizer.
is a modern air sterilizer widely used in custom air cleaners industry. It also enhances the quality custom air cleaners value of the products.
If air sterilizer isn't meeting your needs, or you just want to see what else is out there, check out these content monitoring alternatives Funglan Air Purifier.
It is never too late to have a new mindset and to get things moving in the right direction. Choose Qingdao Funglan Environmental Protection & Technology Co., Ltd. to be your quality provider.
Qingdao Funglan Environmental Protection & Technology Co., Ltd. prepares for every aspect of running a business, and this includes developing a sound understanding and ability to manage the financial aspects of our company, including financial analysis, taxes and budgeting.
The unique connections between air sterilizermanufacturing and customers happen when you find ways to relate on a more personal and engaging level that goes beyond a product.